Pain Remedy PLUS Wireless TENS/EMSCore Products
The Pain Remedy Plus Wireless TENS/EMS System is a battery operated wireless pulse generator that sends electrical impulses from a REMOTE control to a RECEIVER (round) with ELECTRODES attached to it. The Wireless REMOTE controls the INTENSITY and MODE functions of the RECEIVER. The LCD display shows the mode value, channel number, battery life, and intensity level. Because the Pain Remedy Plus System is wireless - needing no lead wires - you must use only Pain Remedy SNAP Electrodes. Lead wire type electrodes will not work with this wireless system.
The Pain Remedy TENS system offers two models
- Pain Remedy Wireless Tens - click here
- Pain Remedy Wireless Tens/EMS (on this page)
They feature a convenient wireless design that can be used almost anywhere on the body. The rechargeable receivers, in combination with snap style electrodes, send electrical impulses to stimulate the nerves and block pain signals. FDA cleared for over the counter sales.
Indications for Use
Temporary relief of pain associated with sore or aching muscles of the back, arms, or legs, due to strain from exercise or normal household and work activities.
The Pain Remedy and Pain Remedy Plus use rechargeable lithium batteries saving you from the hassle and expense of finding AAA batteries when you are in pain. Simply use the charging kit included with each device to recharge the batteries when needed.
Reusable, long-lasting snap electrodes are included with each unit. Replacement electrodes are available for purchase. Select from a 4-pack of 2” x 2” round electrodes or 1-pack of the 3”x5” body electrode.
Use the “tips for skin care” portion of the owners manual to get the most life out of your electrodes.
Receiver Dimensions: 2.25” x .5”
Remote Dimensions: 3.5” x 2.2” x .4”
Power supply:DC 3.7V Lithium
Waveform:Asymmetric biphasic pulseIntensity:0-80 mA @ 500
Pulse Width: 100μS
Frequency: 1- 60 Hz
Mode 1: 53 Hz
Mode 3: 1.3 Hz
Mode 2: 5.8 Hz
Mode 4: 1- 60 Hz
Pulse Amplitude (V): 0-48.5
Mode 1: 0 - 3 7. 9 V
Mode 3: 0-48.5V
Mode 2: 0-46.8V
Mode 4: 0 - 41.1V
Timer: Continuous for 45 minutes duration period; unit automatically shuts off at completion.